Leica upright microscope has motorized objective nosepiece, specimen movement (xy), and focusing (z).
Nitrogen laser has a wavelength of 337nm, a pulse repeat rate up to 30Hz and motorized optical beam scanning and setting of the illumination aperture
Software controls all the motorized microscope functions and laser movements. The smallest cutting area with the 100x objective has a diameter of 4-5µm and an accuracy of +/- 0.6µm
|N Plan||Ph1||10||0.25||5.8||513847||UV||470 LP|
|N Plan||Ph1 correc||20||.040||3.2-1.9||513812||470||515 LP|
|N Plan||Ph2 correc||40||0.55||3.3-1.9||513805||540||590 LP|